Swinburne University of Technology
Our office will be closed Friday, July 3rd in observance of American Independence Day. We will reopen at regular business hours on Monday, July 6th.
ActiGraph's Response to Novel Coronavirus (COVID-19) - June 25th Update
ActiGraph employees are working remotely until July 13th, 2020. Shipping delays should be expected during this time.
Measuring Time in the Office Using Bluetooth Sensors: Feasibility and Validity Considerations
- Published on Mar, 2019
The office is a key setting for intervening to reduce sitting, therefore office-specific activity measures are needed to evaluate interventions. We tested whether valid measures of office time and office-specific activities could be obtained using Bluetooth sensing with a variety of sampling intervals, receiver wear positions, and beacon placements. Workers from one building (n = 29, 72% female, age 23–68 years) wore, for one workday, the activPAL3 on the thigh (measured sitting, standing and stepping) and the Bluetooth-enabled ActiGraph Link on the wrist and thigh. Location (office/not) was estimated by Bluetooth signal presence/absence at two beacons in the wearer’s office (desk, wall), with chest-worn video cameras as the criterion. Accuracy in location classification was assessed and compared across 60-s, 30-s, and 10-s sampling intervals. The validity of Bluetooth-derived measures of total time in the office and in office-specific activities was assessed. For both the wrist and thigh-worn Link, with various beacon placements, accurate classification of location (office/not) was obtained, with a significant (p < .05) but trivial difference in accuracy across sampling interval options (F scores all ≈ .98). With the 60-s sampling interval, mean absolute percent error was very small for office time and office sitting time (<5%), but higher for infrequent activities: standing (17%–23%), incidental stepping (30%–49%), and purposeful walking (57%–86%). The ActiGraph Link can be used to validly measure office time and office location of activity with a 60-s Bluetooth setting. Higher resolution improves accuracy but not to a meaningful degree.
- Bronwyn K. Clark 1
- Nyssa T. Hadgraft 1,2,3
- Takemi Sugiyama 2,4
- Elisabeth A. Winkler 1
The University of Queensland
Baker Heart and Diabetes Institute
Australian Catholic University
Human Kinetics Journals