Our office will remain closed through Friday, September 18th as we continue to assess the damages caused by Hurricane Sally. ActiGraph team members are working remotely, however shipping delays should be expected at this time. We expect to resume regular business hours on Monday, September 21st. If you need immediate assistance, please contact us by email at firstname.lastname@example.org and we will respond as quickly as possible. Thank you for your continued support.
Actigraph GT3X: validation and determination of physical activity intensity cut points.
- Added on April 29, 2014
Abstract: The aims of this study were: to compare energy expenditure (EE) estimated from the existing GT3X accelerometer equations and EE measured with indirect calorimetry; to define new equations for EE estimation with the GT3X in youth, adults and older people; and to define GT3X vector magnitude (VM) cut points allowing to classify PA intensity in the aforementioned age-groups. The study comprised 31 youth, 31 adults and 35 older people. Participants wore the GT3X (setup: 1-s epoch) over their right hip during 6 conditions of 10-min duration each: resting, treadmill walking/running at 3, 5, 7, and 9 km · h⁻¹, and repeated sit-stands (30 times · min⁻¹). The GT3X proved to be a good tool to predict EE in youth and adults (able to discriminate between the aforementioned conditions), but not in the elderly. We defined the following equations: for all age-groups combined, EE (METs)=2.7406+0.00056 · VM activity counts (counts · min⁻¹)-0.008542 · age (years)-0.01380 · body mass (kg); for youth, METs=1.546618+0.000658 · VM activity counts (counts · min⁻¹); for adults, METs=2.8323+0.00054 · VM activity counts (counts · min⁻¹)-0.059123 · body mass (kg)+1.4410 · gender (women=1, men=2); and for the elderly, METs=2.5878+0.00047 · VM activity counts (counts · min⁻¹)-0.6453 · gender (women=1, men=2). Activity counts derived from the VM yielded a more accurate EE estimation than those derived from the Y-axis. The GT3X represents a step forward in triaxial technology estimating EE. However, age-specific equations must be used to ensure the correct use of this device.